You are here: / Home
Low noise and non metallic liquid-helium cryostat
Industry and Research Institutes
Dr. rer. nat. Andreas Kade
Low-noise Magnetic Field Cryostat for SQUID-Applications
ILK Dresden developed low noise magnetic field and non metallic helium cryostats made of GRP (glass reinforced plastics) with a high helium and vacuum holding time.
The cryostats can operate in a direction independent operation mode (tiltable arrangement) and hence they are suitable for a plurality of several applications.
The ILK concept offers much lower permeation rates inside the He-reservoir as comparable models from other manufacturers and is therefore perfect for SQUID (superconducting quantum interference device) applications and other long-term measurements.
The technical design of the cryostats is thus predestined for future sensor generation.
- suitable for the cooling of SQUID-Sensors
- non-metallic (GRP)
- fast filling with LHe via thermosiphon
- GRP exhaust gas cooled radiation shield for high efficiency.
- small cold-warm distance feasible
- low maintenance
- other design variants possible on costumer request
Technical Data for the Cryostats
|volume of liquid helium||5 to 12 litre (other volume possible)|
|time to complete evaporation of helium||3 to 7 days (depends on size)|
|heat load||0.1 W (in standby operation)|
|rate of helium evaporation||≤ 3 litre / day|
|holding time of the liquid helium||up to 4 days without refilling|
|initial noise of the cryostat||< 3 fT / Hz½|
|helium leak test (He-vessel)||< 1 × 10-11 mbar l / s|
Low noise magnetic field
The low-noise magnetic field of the cryostat was tested in the magnetically shielded room BMSR-1 of PTB Berlin and is smaller than the lowest resolution limit of the measurement system used, see figure below.
Elektrochemische Dekontamination leitfähiger Oberflächen „EDeKo II“
Verbesserung der hygienischen Prävention durch elektrochemische Dekontamination
Herstellung neuartiger Sperrschichten an elastomeren Dichtungsmaterialien zur Verminderung der Permeation des Kältemittels R744 (CO2)